- Interventional Cardiology / Radiology / Electrophy
- DSF1533 GORE® DrySeal Flex Introducer Sheath 15Fr., 5.0mm x 5.6mm, 33cm
DSF1533 DrySeal Introducer Sheath 15Fr., Min Sheath ID 5mm, Sheath OD 5.6mmm, Length 33cm Box of 1
The GORE ® DrySeal Flex Introducer Sheath consists of an introducer sheath with the GORE® DrySeal Valve attached, a twist style locking dilator
The introducer sheath is a composite tube which consists of a flat stainless steel wire reinforced hydrophilic coated Pebax® outer tube and PTFE liner with a tapered leading tip and marker band incorporated within the sheath material to allow identification under fluoroscopy. The sheath is attached to the GORE® DrySeal Valve.
The GORE® DrySeal Valve is comprised of an outer silicone tube and an inner film tube. The region between the silicone tube and film tube is pressurized by injecting 2.5 mL of saline into the space, using the provided syringe, during procedural preparation of the device.
The dilator has a tapered leading end and provides dilatation of the access vessel while providing a smooth transition from the guidewire to the introducer sheath leading tip. The dilator is 0.035" guidewire compatible and has a locking mechanism which mates with, and secures to, the Dryseal Valve. The sheath hub is embossed with its French size and a visual marker on the trailing end of the dilator shaft that ensures correct combination of the dilator within the sheath.
• Hydrophilic coated, Flexible Introducer sheath with the GORE® DrySeal Valve attached
• Locking Dilator
• 2.5 mL syringe
The GORE® DrySeal Flex Introducer Sheath is intended to be inserted in the vasculature to provide a conduit for the insertion of endovascular devices while minimizing blood loss associated with such insertions.